# Barometer

Schematic drawing of a simple mercury barometer with vertical mercury column and reservoir at base
Old barometers from the Musée des Arts et Métiers, Paris
Goethe's device
Barometer from the 1890s

A barometer is a scientific instrument used in meteorology to measure atmospheric pressure. Pressure tendency can forecast short term changes in the weather. Numerous measurements of air pressure are used within surface weather analysis to help find surface troughs, high pressure systems and frontal boundaries.

## History

Although Evangelista Torricelli is universally credited with inventing the barometer in 1643,[1][2][3] historical documentation also suggests Gasparo Berti, an Italian mathematician and astronomer, unintentionally built a water barometer sometime between 1640 and 1643.[1][4] French scientist and philosopher René Descartes described the design of an experiment to determine atmospheric pressure as early as 1631, but there is no evidence that he built a working barometer at that time.[1]

On July 27, 1630, Giovanni Battista Baliani wrote a letter to Galileo Galilei explaining an experiment he had made in which a siphon, led over a hill about twenty-one meters high, failed to work. Galileo responded with an explanation of the phenomenon: he proposed that it was the power of a vacuum that held the water up, and at a certain height the amount of water simply became too much and the force could not hold any more, like a cord that can support only so much weight.[5][6] This was a restatement of the theory of horror vacui ("nature abhors a vacuum"), which dates to Aristotle, and which Galileo restated as resistenza del vacuo.

Galileo's ideas reached Rome in December 1638 in his Discorsi. Raffaele Magiotti and Gasparo Berti were excited by these ideas, and decided to seek a better way to attempt to produce a vacuum than with a siphon. Magiotti devised such an experiment, and sometime between 1639 and 1641, Berti (with Magiotti, Athanasius Kircher and Niccolò Zucchi present) carried it out.[6]

Four accounts of Berti's experiment exist, but a simple model of his experiment consisted of filling with water a long tube that had both ends plugged, then standing the tube in a basin already full of water. The bottom end of the tube was opened, and water that had been inside of it poured out into the basin. However, only part of the water in the tube flowed out, and the level of the water inside the tube stayed at an exact level, which happened to be 10.3 m,[citation needed] the same height Baliani and Galileo had observed that was limited by the siphon. What was most important about this experiment was that the lowering water had left a space above it in the tube which had no intermediate contact with air to fill it up. This seemed to suggest the possibility of a vacuum existing in the space above the water.[6]

Torricelli, a friend and student of Galileo, dared to look at the entire problem from a different angle. In a letter to Michelangelo Ricci in 1644 concerning the experiments with the water barometer, he wrote:

Many have said that a vacuum does not exist, others that it does exist in spite of the repugnance of nature and with difficulty; I know of no one who has said that it exists without difficulty and without a resistance from nature. I argued thus: If there can be found a manifest cause from which the resistance can be derived which is felt if we try to make a vacuum, it seems to me foolish to try to attribute to vacuum those operations which follow evidently from some other cause; and so by making some very easy calculations, I found that the cause assigned by me (that is, the weight of the atmosphere) ought by itself alone to offer a greater resistance than it does when we try to produce a vacuum.[7]

It was traditionally thought (especially by the Aristotelians) that the air did not have lateral weight: that is, that the kilometers of air above the surface did not exert any weight above bodies. Even Galileo had accepted the weightlessness of air as a simple truth. Torricelli questioned that assumption, and instead proposed that air had weight, and that it was the latter (not the attracting force of the vacuum) which held (or rather, pushed) up the column of water. He thought that the level the water stayed at (c. 10.3 m) was reflective of the force of the air's weight pushing on it (specifically, pushing on the water in the basin and thus limiting how much water can fall from the tube into it). In other words, he viewed the barometer as a balance, an instrument for measurement (as opposed to merely being an instrument to create a vacuum), and because he was the first to view it this way, he is traditionally considered the inventor of the barometer (in the sense in which we use the term now).[6]

Because of rumors circulating in Torricelli's gossipy Italian neighborhood, which included that he was engaged in some form of sorcery or witchcraft, Torricelli realized he had to keep his experiment secret to avoid the risk of being arrested. He needed to use a liquid that was heavier than water, and from his previous association and suggestions by Galileo, he deduced by using mercury, a shorter tube could be used. With mercury, then called "quicksilver", which is about 14 times heavier than water, a tube only 80 cm was now needed, not 10.5 m.[8]

In 1646, Blaise Pascal along with Pierre Petit, had repeated and perfected Torricelli's experiment after hearing about it from Marin Mersenne, who himself had been shown the experiment by Torricelli toward the end of 1644. Pascal further devised an experiment to test the Aristotelian proposition that it was vapors from the liquid that filled the space in a barometer. His experiment compared water with wine, and since the latter was considered more "spiritous", the Aristotelians expected the wine to stand lower (since more vapors would mean more pushing down on the liquid column). Pascal performed the experiment publicly, inviting the Aristotelians to predict the outcome beforehand. The Aristotelians predicted the wine would stand lower. It did not.[6]

However, Pascal went even further to test the mechanical theory. If, as suspected by mechanical philosophers like Torricelli and Pascal, air had lateral weight, the weight of the air would be less at higher altitudes. Therefore, Pascal wrote to his brother-in-law, Florin Perier, who lived near a mountain called the Puy de Dome, asking him to perform a crucial experiment. Perier was to take a barometer up the Puy de Dome and make measurements along the way of the height of the column of mercury. He was then to compare it to measurements taken at the foot of the mountain to see if those measurements taken higher up were in fact smaller. In September 1648, Perier carefully and meticulously carried out the experiment, and found that Pascal's predictions had been correct. The mercury barometer stood lower the higher one went.[6]

## Types

### Water-based barometers

The concept that decreasing atmospheric pressure predicts stormy weather, postulated by Lucien Vidi, provides the theoretical basis for a weather prediction device called a "storm glass" or a "Goethe barometer" (named for Johann Wolfgang Von Goethe, the renowned German writer and polymath who developed a simple but effective weather ball barometer using the principles developed by Torricelli).

The weather ball barometer consists of a glass container with a sealed body, half filled with water. A narrow spout connects to the body below the water level and rises above the water level. The narrow spout is open to the atmosphere. When the air pressure is lower than it was at the time the body was sealed, the water level in the spout will rise above the water level in the body; when the air pressure is higher, the water level in the spout will drop below the water level in the body. A variation of this type of barometer can be easily made at home.[9]

### Mercury barometers

A mercury barometer has a glass tube with a height of at least 84 cm, closed at one end, with an open mercury-filled reservoir at the base. The weight of the mercury creates a vacuum in the top of the tube. Mercury in the tube adjusts until the weight of the mercury column balances the atmospheric force exerted on the reservoir. High atmospheric pressure places more force on the reservoir, forcing mercury higher in the column. Low pressure allows the mercury to drop to a lower level in the column by lowering the force placed on the reservoir. Since higher temperature at the instrument will reduce the density of the mercury, the scale for reading the height of the mercury is adjusted to compensate for this effect.

Torricelli documented that the height of the mercury in a barometer changed slightly each day and concluded that this was due to the changing pressure in the atmosphere.[1] He wrote: "We live submerged at the bottom of an ocean of elementary air, which is known by incontestable experiments to have weight".[10]

The mercury barometer's design gives rise to the expression of atmospheric pressure in inches or millimeters or feet (torr): the pressure is quoted as the level of the mercury's height in the vertical column. Typically, atmospheric pressure is measured between 26.5 to 31.5 in Hg. One atmosphere (1 atm) is equivalent to 760 millimeters of mercury.

Design changes to make the instrument more sensitive, simpler to read, and easier to transport resulted in variations such as the basin, siphon, wheel, cistern, Fortin, multiple folded, stereometric, and balance barometers. Fitzroy barometers combine the standard mercury barometer with a thermometer, as well as a guide of how to interpret pressure changes. Fortin barometers use a variable displacement mercury cistern, usually constructed with a thumbscrew pressing on a leather diaphragm bottom. This compensates for displacement of mercury in the column with varying pressure. To use a Fortin barometer, the level of mercury is set to the zero level before the pressure is read on the column. Some models also employ a valve for closing the cistern, enabling the mercury column to be forced to the top of the column for transport. This prevents water-hammer damage to the column in transit.

On June 5, 2007, a European Union directive was enacted to restrict the sale of mercury, thus effectively ending the production of new mercury barometers in Europe.

### Vacuum pump oil barometer

Using vacuum pump oil as the working fluid in a barometer has led to the creation of the new "World's Tallest Barometer" in February 2013. The barometer at Portland State University (PSU) uses doubly distilled vacuum pump oil and has a nominal height of ~12.4 m for the oil column height; expected excursions are in the range of ±0.4 m over the course of a year. Vacuum pump oil has very low vapor pressure and it is available in a range of densities; the lowest density vacuum oil was chosen for the PSU barometer to maximize the oil column height.[11]

### Aneroid barometers

Old aneroid barometer
Modern aneroid barometer

An aneroid barometer is an instrument for measuring pressure as a method that does not involve liquid. Invented in 1844 by French scientist Lucien Vidi[12] uses a small, flexible metal box called an aneroid cell (capsule), which is made from an alloy of beryllium and copper. The evacuated capsule (or usually more capsules) is prevented from collapsing by a strong spring. Small changes in external air pressure cause the cell to expand or contract. This expansion and contraction drives mechanical levers such that the tiny movements of the capsule are amplified and displayed on the face of the aneroid barometer. Many models include a manually set needle which is used to mark the current measurement so a change can be seen. In addition, the mechanism is made deliberately "stiff" so that tapping the barometer reveals whether the pressure is rising or falling as the pointer moves. This type of barometer is common in homes and in recreational boats but also used in meteorology, mostly as barograph and pressure instruments in radiosondes.

### Barographs

A barograph records a graph of some atmospheric pressure and uses an aneroid barometer mechanism to move a needle on a smoked foil or to move a pen upon paper, both of which are attached to a drum moved by clockwork.[13]

### Mems Barometers

Microelectromechanical systems (or Mems) barometers are extremely small devices between 1 to 100 micrometres in size (i.e. 0.001 to 0.1 mm). They are created via lithography or etching. Typical applications include miniaturized weather stations, electronic barometers and altimeters.[14]

### More unusual barometers

The Galaxy Nexus has a built-in barometer[15]

There are many other more unusual types of barometer. From variations on the storm barometer, such as the Collins Patent Table Barometer, to more traditional looking designs such as Hooke's Otheometer and the Ross Sympiesometer. Some, such as the Shark Oil barometer,[16] work only in a certain temperature range, achieved in warmer climates.

A barometer can also be found in smartphones such as the Samsung Galaxy Nexus smartphone,[17] which is included to provide a faster GPS lock.[18]

## Applications

Digital graphing barometer.
Analogue recording Barograph using five stacked aneroid barometer cells.

Using barometric pressure and the pressure tendency (the change of pressure over time) has been used in weather forecasting since the late 19th century.[19] When used in combination with wind observations, reasonably accurate short-term forecasts can be made.[20] Simultaneous barometric readings from across a network of weather stations allow maps of air pressure to be produced, which were the first form of the modern weather map when created in the 19th century. Isobars, lines of equal pressure, when drawn on such a map, gives a contour map showing areas of high and low pressure.[21] Localized high atmospheric pressure acts as a barrier to approaching weather systems, diverting their course. Atmospheric lift caused by low-level wind convergence into the surface low brings clouds and potentially precipitation.[22] The larger the change in pressure, especially if more than 3.5 hPa, the larger the change in weather can be expected. If the pressure drop is rapid, a low pressure system is approaching, and there is a greater chance of rain . Rapid pressure rises, such as in the wake of a cold front, are associated with improving weather conditions, such as clearing skies.[23]

## Compensations

### Temperature

The density of mercury will change with temperature, so a reading must be adjusted for the temperature of the instrument. For this purpose a mercury thermometer is usually mounted on the instrument. Temperature compensation of an aneroid barometer is accomplished by including a bi-metal element in the mechanical linkages. Aneroid barometers sold for domestic use typically have no compensation under the assumption that they will be used within a controlled room temperature range.

### Altitude

As the air pressure will be decreased at altitudes above sea level (and increased below sea level) the uncorrected reading of the barometer will be dependent upon its location. This pressure measurement is then converted to an equivalent sea-level pressure for purposes of reporting. For example, if a barometer located at sea level is moved to an altitude of 1,000 feet (305 m), about 1 inch of mercury (~3.4 kPa) is added on to the reading. The barometer readings at the two locations should be the same if there are negligible changes in time, horizontal distance, and temperature.

Aneroid barometers have a mechanical adjustment for altitude that allows the equivalent sea level pressure to be read directly and without further adjustment if the instrument is not moved to a different altitude. Setting an aneroid barometer is similar to setting an analog clock that is not at the correct time. Its dial is rotated so that the current atmospheric pressure from a known accurate and nearby barometer (such as the local weather station) is displayed. Though somewhat rare, a few aneroid barometers intended for monitoring the weather are calibrated to manually adjust for altitude. In this case, knowing either the altitude or the current atmospheric pressure would be sufficient for future accurate readings.

Aircraft altimeters may need to be adjusted as they fly between regions of varying normalized atmospheric pressure, due to the presence of weather systems.

## Barometers and atmospheric pressure calculations

[24]

When atmospheric pressure is measured by a barometer, the pressure is also referred to as the barometric pressure. Let's take a barometer with a cross-sectional area, A, a height, h, filled with mercury from the bottom at Point B to the top at Point C. The pressure at the bottom of the barometer, Point B, is equal to the atmospheric pressure. The pressure at the very top, Point C, can be taken as zero because there is only mercury vapor above this point and its pressure is very low relative to the atmospheric pressure. Therefore, one can find the atmospheric pressure using the barometer and this equation:

Patm = ρgh

where ρ is the density of mercury, g is the gravitational acceleration, and h is the height of the mercury column above the free surface area. Note that the physical dimensions (length of tube and cross-sectional area of the tube) of the barometer itself have no effect on the height of the fluid column in the tube.

In thermodynamic calculations, a commonly used pressure unit is the standard atmosphere. This is the pressure resulting from a column of mercury of 760mm in height at 0 °C. For the density of mercury, use ρHg = 13,595 kg/m3 and for gravitational acceleration use g = 9.807 m/s2.

If water were used (instead of mercury) to meet the standard atmospheric pressure, a water column of roughly 10.3 m (33.8 ft) would be needed.

Standard atmospheric pressure as a function of elevation.

Note: 1 torr = 133.3 Pa = 0.03937 In Hg

PatmAltitudePatmAltitude
101.325 kPaSea Level (0m)29.92 In HgSea Level (0 ft)
97.71 kPa305 m28.86 In Hg1,000 ft
kPa610 mIn Hg2,000 ft
89.88 kPa1,000 m26.55 In Hg3,281 ft
84.31 kPa1,524 m24.90 In Hg5,000 ft
79.50 kPa2,000 m23.48 In Hg6,562 ft
69.68 kPa3,048 m20.58 In Hg10,000 ft
54.05 kPa5,000 m15.96 In Hg16,404 ft
kPa6,096 mIn Hg20,000 ft
37.65 kPa7,620 m11.12 In Hg25,000 ft
kPa10,000 mIn Hg32,808 ft
11.65 kPa15,240 m3.44 In Hg50,000 ft
5.53 kPa20,000 m1.63 In Hg65,617 ft

## Patents

Table of Pneumaticks, 1728 Cyclopaedia

## References

1. ^ a b c d "The Invention of the Barometer". Islandnet.com. Retrieved 2010-02-04.
2. ^ "History of the Barometer". Barometerfair.com. Retrieved 2010-02-04.
3. ^ "Evangelista Torricelli, The Invention of the Barometer". Juliantrubin.com. Archived from the original on 9 February 2010. Retrieved 2010-02-04.
4. ^ Drake, Stillman (1970). "Berti, Gasparo". Dictionary of Scientific Biography 2. New York: Charles Scribner's Sons. pp. 83–84. ISBN 0-684-10114-9.
5. ^ William R. Shea (2003). Designing Experiments & Games of Chance: The Unconventional Science of Blaise Pascal. Science History Publications. pp. 21–. ISBN 978-0-88135-376-1. Retrieved 10 October 2012.
6. "History of the Barometer". Strange-loops.com. 2002-01-21. Archived from the original on 6 January 2010. Retrieved 2010-02-04.
7. ^ "Torricelli's letter to Michelangelo Ricci". Web.lemoyne.edu. Retrieved 2010-02-04.
8. ^ "Brief History of the Barometer". Barometer.ws. Archived from the original on 14 January 2010. Retrieved 2010-02-04.
9. ^ Jet Stream. Learning Lesson: Measure the Pressure – The "Wet" Barometer. Retrieved on 2007-05-05.
10. ^ Strangeways, Ian. Measuring the Natural Environment. Cambridge University Press, 2000, p. 92.
11. ^ Tomlinson, Stuart (February 10, 2013) Large barometer at Portland State University could be the tallest in the world. oregonlive.com
12. ^ Louis Figuier; Émile Gautier (1867). L'Année scientifique et industrielle. L. Hachette et cie. pp. 485–486.
13. ^ Glossary of Meteorology. Barograph. Retrieved on 2007-05-05.
14. ^ Sensors & Transducers e-Digest, Vol. 92, Issue 4, May 2008. "MEMS Barometric Pressure Sensor". sensosportal.com. International Frequency Sensor Association (IFSA). Retrieved 13 June 2014.
15. ^ Galaxy Nexus. Google.com. Retrieved on 2011-12-03.
16. ^ Shark Oil Barometer Barometer World Retrieved on 2009-09-26.
17. ^ This Is the Samsung Galaxy Nexus, Google's New Official Android Phone. Gizmodo.com (2011-10-18). Retrieved on 2011-11-15.
18. ^ Galaxy Nexus barometer explained, Sam Champion not out of a job. Engadget (2011-10-20). Retrieved on 2011-12-03.
19. ^ USA Today. Understanding air pressure. Retrieved on 2008-05-25.
20. ^ USA Today. Using winds and a barometer to make forecasts. Retrieved on 2007-05-05.
21. ^ Edward J. Hopkins, Ph.D. (1996-06-10). "Surface Weather Analysis Chart". University of Wisconsin. Archived from the original on 28 April 2007. Retrieved 2007-05-10.
22. ^ Robert Penrose Pearce (2002). Meteorology at the Millennium. Academic Press. p. 66. ISBN 978-0-12-548035-2. Retrieved 2009-01-02.
23. ^ Weather Doctor. Applying The Barometer To Weather Watching. Retrieved on 2008-05-25.
24. ^ Thermodynamics: An Engineering Approach by Yunus A. Cengal and Michael A. Boles